From: Alberto Fasso' (fasso@SLAC.Stanford.EDU)
Date: Mon Mar 03 2008 - 02:28:10 CET
the detection efficiency is the results of several different things.
Some of them FLUKA can calculate, but some information must be provided
1) Geometrical acceptance of the detector: FLUKA does it
2) Energy deposited in the detector: FLUKA does it
3) Response of the detector for that deposited energy: needs some input
For point 3) there are two possibilities:
a) to provide one or two Birks quenching coefficients (see the manual
about command TCQUENCH). For some scintillators, these coefficients have
been measured and published, see for instance the reference mentioned
in the example of the manual following the description of the command.
This choice only gives you the fraction of deposited energy which is
converted to scintillation photons, but not the self-absorption of those
photons inside the detector, nor the fraction of photons reaching the
photocathode and being detected by it.
b) to simulate explicitely the production of scintillation photons (see
command OPT-PROD, with SDUM = SCINTILL or SCINT-WV or SCINT-OM), and their
transport (command OPT-PROP).
You must provide up to 3 photon monochromatic energies, together with the
respective fraction of deposited energy going into each of the 3 lines.
You must also provide the optical properties of the scintillator (energy
or wavelength range for the transport, refraction and absorption index)
and the quantum efficiency, or sensitivity of the the photocathode.
As the latin poet Catullus said:
Coenabis bene, mi Fabulle, apud me
si tecum attuleris bonam atque magnam
coenam, non sine candida puella...
(You will dine well at my table, dear Fabullus,
if you will bring with you a good and big dinner,
and possibly also a nice girl...)
On Wed, 27 Feb 2008, Sandra Moretto wrote:
> Dear fluka-users,
> my situation is very simple.
> I have an isotropic source of neutrons with E<20 MeV.
> and a Cylindric Scintillator placed at 50 cm away from the source.
> I'd like to calculate the detection efficiency.
> It is feasible to use fluka for that?
> Thanks for the attention
> SANDRA MORETTO
> Dipartimento di Fisica
> Via Marzolo 8 - 35131
> Padova (ITALY)
> Tel: + 39 049-827 7181
> E-mail: firstname.lastname@example.org
-- Alberto Fasso` SLAC-RP, MS 48, 2575 Sand Hill Road, Menlo Park CA 94025 Phone: (1 650) 926 4762 Fax: (1 650) 926 3569 email@example.com
This archive was generated by hypermail 2.1.6 : Mon Mar 03 2008 - 09:22:05 CET